COMETEO Professional Multi-Plate Radiation Shield for Weather Sensors
Professional passive solar radiation shield for better protection of weather sensor, datalogger and barometers, naturally ventilated
"Not only radiation shield, but the complete COMETEO system minimizing negative effects"
Multiplate radiation shield is used to protect measuring devices and provides more accurate measurement results. The shield minimizes radiation reaching the sensor, minimizes radiation absorbed by the shield and maximizes ambient air flow around the sensor. The enlarged top plate „cap“ is 210mm in diameter and designed to provide shade for high solar angles for lower plates and also for full protection of measuring device placed underneath.
Works with COMET instruments - see OPTIONAL ACCESSORIES
Request your free COMETEO shield for six months evaluation.
If you are satisfied after testing you can buy the shield, otherwise return it.
GENERAL TECHNICAL DATA
Air Temperature in Meteorology
The measurement of air temperature in meteorology is a difficult task. Accurate air temperature measurements require minimization of local effects (e.g. trees and buildings) and minimization of environmental effects that influence the actual sensing of the temperature at the position of the sensor. World Meteorological Organization - WMO defines internationally the temperature of the air nearby the earth's surface as “the temperature indicated by a thermometer exposed to the air in a place sheltered from direct solar radiation” (WMO, 1992).
The main functions of thermometer shields are to protect the sensor from direct or indirect radiation from the sun during the day and from radiation from the sensor to the sky at night and to protect the sensor from wetting. Wetting of the sensor and screen is caused by fog, drizzle or rain. But protection against radiation and wetting conflicts with the requirement of sufficient ventilation. Due to insufficient ventilation a microclimate develops within the screen. In fact, all screens develop their own microclimate and the difference with the ambient climate depends on screen type and design. It may be stated that the larger the bulk of a screen, the stronger the microclimate within the screen and the more the sensed temperature deviates from the real air temperature. From the above, it follows that thermometer screens should have a high reflectivity to minimize heating of the plates and subsequent warming of the air as it flows over the plates to the sensor. To decrease the time the air is in contact with the plate, the plate radial distance should be as small as possible. To prevent the development of microclimates within screens, also the blockage of ambient airflow by the screen should be minimized by making the plate spacing as large as possible, taking into account the requirement that wetting of the sensor should be prevented. Any design of a thermometer screen is a result of compromises. Finding the optimal design is a serious challenge that may require the use of different methods for studying the behavior of screens. In the field, the actual differences between the air temperature in the screens and the ambient air temperature depend mainly on the prevailing weather conditions. Furthermore, the state of the ground and humidity of the soil will also affect the air temperature in the screen as radiation or reflection will cause screen dependent biases. This can easily occur in areas with snow cover and in deserts.
Meteorology - COMET sensors together with COMET radiation shields are the ideal basis for weather stations for long-term weather, climate and air monitoring.
Building management - Our instruments which measure temperature, humidity, atmospheric pressure and CO2 are an integral part of these industry-leading solutions.
COMETEO Professional Radiation Shield for Weather Sensors, actively ventilated with built-in fan
COMETEO Professional Universal Multi-plate Radiation Shield for Weather Sensors
Temperature and humidity duct probe with 4-20mA output
Temperature and humidity duct probe with 0-10V output
Temperature and humidity duct mount transmitter with RS485 output
Web Sensor with PoE, duct mount - remote thermometer hygrometer barometer with Ethernet interface
Temperature and humidity probe with 0-10V output
RH+T+Tdp cable probe, output 4-20mA
Industrial temperature, humidity, bar. pressure transmitter - RS232 output
Industrial temperature, humidity, bar. pressure transmitter - RS485 output
Web Sensor - remote thermometer hygrometer barometer with Ethernet interface
Duct mount temperature probe -30 to +80°C / 4 to 20mA, IP65
Duct mount temperature transmitter with 0-10V output, stem length 120mm
Temperature transmitter with 4-20mA output
Temperature transmitter with 0-10V output
Temperature transmitter with RS232 interface
Temperature transmitter with RS485 interface
WebSensor - remote thermometer with Ethernet interface
Temperature Transmitter Pt1000 -30°C to +80°C/ 4 to 20 mA
Thermo-hygrometer with external probe, cable 1 meter
Thermo-hygro-barometer with external probe
Ethernet Multilogger - thermo hygro meter with 4 MiniDIN inputs
GSM temperature and humidity data logger for external probe with built-in modem